Bob Beck Protocol - Blood Purifier
Stage IV Protocol
This article is only a small part of the complete Bob Beck Protocol article. Do NOT read this article without reading the complete Bob Beck Protocol article first:
Article #1: Blood Purifier or Blood Electrification
The purpose of the Blood Purifier/Blood Electrification is to attenuate (i.e. disable from multiplying) microbes which flow through the blood. These microbes die without ever being replaced.
The Blood Purifier creates a small electrical current in the blood which will break off a key enzyme on the surface of all microbes which are traveling through the blood. This prevents the microbes from entering human cells and multiplying therein. The microbes are then safely removed from the body. The electrical current can also outright kill bacteria and other pathogenic invaders.
Because of the vast number of deactivated viruses and dead microbes it is almost certain that within the first few days of using the Blood Electrification device you will experience "flu-like symptoms" (this is called the "Herxheimer Reaction"). These symptoms are caused by the massive die-off of microbes and do not create any type of danger.
To prevent this symptom from being severe, it is important to drink copious amounts of water during the Bob Beck Protocol. This water includes the ozonated water.
The rule of thumb is to drink half your body weight (measured in pounds) in water (measured in ounces). For example, if your weight is 200 pounds, you would drink 100 ounces of water (1/2 of 200 is 100) per day, maybe more. This includes the ozonated water.
Because the Magnetic Pulser is generally used prior to the Blood Electrification treatment, it is important to understand when the water should be taken:
You also might want to drink some water immediately after using the Magnetic Pulser and wait 15 to 30 minutes before starting this Blood Purifier treatment. This water is to detoxify (especially the ozonated water) the blood and lymph system.
In other words, the majority of the water you drink during the day should be at one of those three times. All three times are before the Blood Purifier. Most, if not all, of the water you take should be ozonated water.
If you think you have large amounts of microbes in your body at the beginning of this treatment, which most cancer patients do have, you might want to use the Blood Purifier for only 30 minutes on the first day, and then build up by 15 minutes a day, until you are using the machine for two hours. Within one week you should be up to the full two hour protocol.
However, if a cancer patient is very advanced, and is not expected to live more than two months, then do NOT do the buildup. In other words, start at two hours on the first day.
While this protocol may be used until the cancer is cured, in fact this protocol should be used for life. This is not because of cancer, but for overall health. Two pounds of microbes in a person's bloodstream is not healthy for anyone.
There are two different vendors of the Bob Beck Blood Purifier. Their equipment is very different and the instructions for their use are very different. The two vendors are Sota Instruments and BioElecrtic.
The instructions for using the Sota are in the next section. The instructions for using the BioElectric will be linked to below in the section on BioElectric. BioElectric devices generally arrive within two weeks.
Also, the Sota instrument follows the original Bob Beck specifications. The BioElectric brand uses a different philosophy. My own experiments on the BioElectric is that it clearly gets rid of the microbes.
The BioElectric device does have some advantages. It can also be used on other health conditions such as sinusitis because of your flexibility of placing the electrodes. It would also likely do a better job on curing hepatitis C if the electrodes were placed such that the electrical current flowed directly through the liver. Between the electrodes the electrical current would look like a football, if you could see it.
Having said all of that, we know the Sota works and we are confident, based on our own research, that the BioElectric will work.
Analysis of Sota Instruments Blood Purifier
VERY, VERY Important Note - Study Carefully!!!:
When placing the electrodes on your ulnar and radial arteries (on your wrists), you should place the electrodes directly on top of the arteries and at the location of strongest pulse. Once you get used to placing the electrodes you will memorize their location and be familiar with the amount of tingling of the electrical current.
Before putting the electrodes on your wrist, each metal electrode should be placed inside of a cotton sleeve. The purpose of these cotton sleeves is to protect your skin and to provide better electrical conductivity between the electrodes and the skin. The sleeves need to be completely soaked in tap water (or sea salt water if the tap water isn't very electrically conductive) in order for the electricity to be transferred from the electrodes to the skin.
The following instructions are for use of blood electrifiers (Sota, etc) that only allow the two electrodes to be placed on one wrist. This is not as easy to do as when the two electrifiers are placed one on each wrist (i.e. the way BioElectric does it). With single wrist placement too much electrical current just crosses the wrist instead of entering the blood in the arteries. To offset that so much current stays at the level of the skin a higher amount of electrical current needs to be applied. Some people with delicate skin can't tolerate this level of current. Overweight people also may have problems finding both arteries.
When keeping the electrode sleeves wet, by putting a few drops of water on them every 20 to 60 minutes, it is possible that there will be a film of water on the surface of the wrist, and/or on the bottom of the rubber strap, between the two cotton sleeves. Since electricity always takes the path of least resistance, it is possible the electricity will cross over the thin layer of water on your skin or the bottom of the rubber strap, instead of going through the skin and into the blood stream. You will know this happens because the tingling of the skin, caused by the electrical current, will stop after you put drops of salt water on the cotton sleeves.
It is critical to keep the skin between the two cotton sleeves, and the bottom of the rubber strap, DRY. This is very important to check after each time you wet the cotton sleeves. You can use a Q-Tip, and/or a dry finger and/or blow air between the electrodes, until the area between the cotton sleeves is dry!! When the area between them is dry (including the top of the skin and the bottom of the rubber strap), the tingling should return.
I am not an electrical engineer, so when I saw how much more complicated the electronics of the Sota Instruments Blood Purifier was, compared to the original Bob Beck schematics, I decided to have a real electrical engineer look at it to see if it was true to the original. All of the things I learned in dealing with this guru are worth noting and there are some surprises!! The second item is the most important!
First, after a great deal of effort on my part, and on the part of an electronics genius (who designs equipment most people couldn't even operate), his conclusion was that the Sota Instruments Blood Purifier WAS true to the original. He used his lab equipment to analyze the actual output of the Sota Blood Purifier and he studied the schematics of the original Beck equipment and compared it to the actual Sota circuits. It was all thumbs up.
Second, the dial on the Blood Purifier controls the flow of electrical current. In at least one of the Bob Beck videos, available on Google, Bob Beck tells the audience to adjust the dial to the level of comfort. While this is true, he forgot to mention a very important thing!
During the first half hour, every 5 or 10 minutes you need to turn the knob closer to the maximum. You may be able to turn it up slightly 2 or 3 times during this half hour. In other words, every 5 or 10 minutes you may be able to tolerate a little more tingling. If it is very uncomfortable then back off on the dial just a little bit.
Some people, perhaps most people, may not be able to tolerate the maximum dial setting. Just do the best you can without being too uncomfortable.
The bottom line is this, you should adjust the Blood Purifier to a "slightly uncomfortable" position. Do not adjust it to an uncomfortable position, just to a "slightly uncomfortable" position.
Third, he said there was no reason a person could not build the Blood Electrifier at home using the original Beck schematic (which is linked to in the main article).
Analysis of BioElectric's "Microbe Electrifier"
Michael Forrest's BioElectric "Microbe Electrifier" is the only one that also allows you to use frequencies higher than Beck's 4 hertz in order to avoid the transfection/electroporation effect. He relies on a graph of frequency verses transfection in the Biophysical Journal, volume 58, 1990 as his proof that 10 hertz and above cause no transfection.
He says that although Beck settled on 3.9 hertz, he had admitted that the deactivation of microbes was not dependant on a specific frequency. The tests by Lyman and Kaali were actually with zero frequency- direct current! But DC causes blood cell clumping as well as transfection. Use of 10 or 40 hertz can be important for cancer patients if they want to continue to use strong herbs or other strong supplements.
BioElectric also recommends placing the electrodes wrist-to-wrist or ankle-to-ankle in order to be able to use comfortable levels of electrical current because most of the current will concentrate itself in the arteries between the shoulders and in the groin area.
Other companies, such as Sota Instruments, recommend placing electrodes on one wrist because that is the last recommendation Beck gave. But BioElectric says Beck did that only for convenience sake and that Beck's own research proved that 97% of the current was lost by it crossing the wrist instead of going into the arteries!!
Beck admitted that the extreme amount of 3mA (3 milliamps) had to be applied wrist to wrist in order for the correct amount of current to be measurable in the forearm artery. BioElectric says that very few people can tolerate that amount of electricity. You can read more at his:
Using electrodes wrist to wrist or ankle to ankle only requires a moderate amount of current in the range of .13mA to .25mA as is explained in his blood current page. This amount is tolerable by everyone as it poses no threat of skin irritation. The Microbe Electrifier has a light that comes on when the user has adjusted the current control to acheive a current level within this range.
BioElectric's Microbe Electrifier is definitely off the beaten path of all others who stick strictly to Becks last recommendations but he backs up all his improvements with research and logic. His philosophy is that all original ideas can and should be improved upon and that we shouldn't limit ourselves to original designs. If we did we would all still be driving model T Fords.
WARNING #1: Whichever blood purifier you select you need to remember to apply water to the cloth/pads (i.e. wet it down) every 30 minutes!! If the pads get dry they will not transmit electricity!!
WARNING #2: Whichever blood purifier you select you should start the session by turning the dial up until the electrical current is noticeable and then turn it slightly to the left until it is no longer noticeable. There are several reasons for doing this. First, if you forgot to add baking soda to the water, this will catch that error. Second, this is the most effective setting for the device to attenuate microbes. Third, if you forgot to turn your "on" switch to "on," it will catch that error also.